Surgical Outcomes Of Adjacent Segment Disease After Lumbar Fusion Created by host on 10/31/2011 12:00:46 AM
Radiographic changes of adjacent segment deterioration after lumbar fusion have commonly been reported in the literature. Symptomatic adjacent segment disease (ASD) requiring surgical treatment is less common.
OF SURGICAL TREATMENT OF ADJACENT SEGMENT DISEASE
AFTER LUMBAR FUSION WITH LUMBAR I/F CAGES AND
THE VARIABLE SCREW PLACEMENT SYSTEM
Lesley Shure M.D.* Guy R. Fogel M.D.** John S. Toohey M.D. Arvo Neidre M.D. South Texas Orthopaedic and Spinal Surgery Associates 9150 Huebner Road San Antonio Texas, 78240 (210)561-7234 FAX: (210)561-7240 e-mail: email@example.com
Device status: The Lumbar Interbody carbon fiber Cage and the VSP Spinal Fixation System are approved by the United States Food and Drug Administration.
* DePuy Spine sponsored spine surgery fellowship. **Corresponding author: Dr. Guy R. Fogel No financial assistance was provided from any source for this study.
BACKGROUND CONTEXT: Radiographic changes of adjacent segment deterioration after lumbar fusion have commonly been reported in the literature. Symptomatic adjacent segment disease (ASD) requiring surgical treatment is less common. Clinical management of ASD is difficult, and in some cases surgery is indicated. The surgical treatment of adjacent segment stenosis or instability after lumbar fusion has been seldom addressed. In this study, posterior lumbar interbody fusion (PLIF) is applied as surgical treatment for symptomatic lumbar ASD. PURPOSE: To report functional outcomes, radiographic fusion rates, further need for re-operation in the surgical treatment of ASD. STUDY DESIGN/SETTING: This retrospective, single-arm cohort study of a community surgical practice evaluates ASD patients treated with PLIF. PATIENT SAMPLE: Between 1999 and 2002, 44 patients underwent PLIF at an adjacent segment following a previous lumbar fusion. At more than three years, 34(77%) were available for follow-up. OUTCOME MEASURES: Clinical outcomes were based on the modified Prolo score, evaluating pain, economic status, function and medication usage. Fusion success, on plain radiographs, was defined by continuous bone bridging the interbody and posterior-lateral fusion level, with no interrupting lucencies, on plain radiographs. The worst remaining adjacent level at the time of the index surgery was graded and compared to last follow-up. Lumbar lordosis, sagittal and coronal alignment was measured. METHODS: Forty-four patients with previous lumbar fusion underwent second lumbar spine surgery for adjacent instability. All were treated with autogenous interbody and posterior-lateral arthrodesis and pedicle screw fixation in addition to decompressive laminectomy. Medical records and radiographs were reviewed, and final Prolo scores were obtained.
RESULTS: The clinical success was 88.2% with 55% rated as fair. Radiographic fusion success was 91.2%. After augmentation of the posterior-lateral fusion in three patients the fusion rate ultimately was 100%. One had revision of a displaced interbody cage. 18% had extension of fusion to adjacent levels. 43% had pedicle screw removal. 9% had non-device related complication including dural tear in one, post-operative seroma in two and one deep infection. The UCLA grades identified the worst adjacent degenerative level before the index adjacent level fusion averaged 1.3 and at last follow-up was average 2.4 with an average change of 1. The lumbar lordosis averaged 45 degrees (18-78 degrees). There were 14 (32%) with final lordosis below 40 degrees (average 30 degrees (18-39 degrees). The sagittal alignment was mildly abnormal in three (1.5 cm off plumb) and greater than 20 degrees scoliosis angulation from L1-S1 in two patients.
CONCLUSIONS: Adjacent segment disease may require surgical treatment when indicated for severe symptoms and loss of daily functional activities. Symptomatic ASD may be best treated with PLIF. When compared to previous series of treatment for ASD, PLIF gives comparable rates of clinical success and fusion success. When compared to our previous experience clinical success has more fair and poor results in ASD, and there is a statistically significant lower rate of fusion success in ASD.
PLIF (Posterior Lumbar Interbody Fusion) PLF (Posterior-lateral Fusion) Interbody fusion orthosis Cage Brantigan cage Pedicle screw Steffee plate Adjacent segment degeneration Adjacent segment disease
Intervertebral Disk/pathology/surgery Lumbar Vertebrae/pathology/radiography/surgery Postoperative Complications Spinal Fusion/*instrumentation/*methods Transitional Disease Treatment Outcome
Adjacent segment deterioration is commonly described after lumbar fusion. Adjacent segment deterioration includes the adjacent or second adjacent motion segment. The more common findings include disc degeneration, hypertrophic changes, multifactorial spinal stenosis; and instability. Adjacent segment disease (ASD) is the clinical syndrome characterized by Hillibrand as the development of new clinical symptoms correlating to the radiographic adjacent segment deterioration20. Although the development of adjacent segment deterioration is be a part of the normal aging process, this process may be accelerated by the altered mechanics that occur with lumbar fusion. Long-term studies of scoliosis and more limited lumbar fusion have suggested fusion increases degenerative changes at adjacent motion segments13,16,20,26,28,30,35,38. This radiographic deterioration may not be symptomatic13,21,32. In fact, in most series functional outcomes are largely unaffected by radiographic adjacent segment deterioration2,13,14,16,23,28,30,32.
Surgical treatment of adjacent segment disease is not often required. Surgery is considered in ASD patients with radiographic deterioration and symptoms of back pain usually accompanied by radicular leg pain, not relieved by conservative measures. Surgical treatment of ASD is a form of revision spine surgery. The overall clinical success of revision spine surgery ranges from 60-80%12,22. The rate of fusion success in revision surgery is also less than primary results. This report identified 44 patients who had posterior lumbar interbody fusion (PLIF) at adjacent levels to a previous lumbar fusion for the indication of ASD. The purpose of this retrospective review is to report the clinical success, fusion success, need for further surgery, and complications of the surgical treatment of ASD.
Materials and Methods
In a single surgical practice from 1999 to 2002, 44 patients were identified by chart review as having undergone PLIF specifically for treatment of ASD. All had disabling back and/or radicular pain refractory to conservative management associated with adjacent segment deterioration on imaging. Demographic data were recorded, including age, gender, number and type of prior back surgeries, diagnosis, surgical levels, complications, re-operations including revisions. The Lumbar I/F cage and the Variable Screw Placement System have been previously described 7. The surgical technique features autograft filled interbody cages and posterior-lateral fusion with pedicle screw fixation6,7. Exploration of adjacent previous fusion was done and if stable, hardware was removed.
Standard biplanar x-rays of the spine with flexion and extension views were reviewed of the preoperative and last visit for each patient. The lateral x-ray was measured for inter-vertebral disc height, the anterior-posterior translation and angular motion of the vertebral bodies. Sagittal balance and lumbar lordosis were recorded. Radiographic evidence of instability was defined, as described by Wiltse and Winter, as >4 mm of translation or >10° of angular motion between adjacent end plates on lateral flexion and extension radiographs when compared with the adjacent cephalic and caudal levels54. A degenerative grade was assigned at each lumbar disc level at the pre-operative and last visit. The amount of lumbar degeneration was classified according to the University of California at Los Angeles grading scale as described by Ghiselli et al (Table 1) 15. The worst adjacent segment was identified before the index surgery and followed to the last available radiograph. Clinical success was defined according to previous published literature parameters used over many years and modeled after an expanded Prolo scale6,7,40. The 5-point Likert scales for pain, function, economic status, and medication usage are added to a combined 4- and 20-point scale33. This study defined a patient’s result as a clinical success when the final rating was excellent (17 to 20 points) or good (13 to 16 points), or fair (9 to 12 points) if the patient achieved a minimum improvement of three points or more in the combined 20-point scale. Interviews of the patients by telephone were done, in order to obtain their final Prolo scores. At the time of the original Brantigan carbon fiber cage report in 1990, there was little agreement regarding optimum outcome measures in a spinal fusion study. To our knowledge, the Prolo Scale has not had a thorough examination of its statistical validity. In the original IDE and the ten-year follow-up, it has been shown that the Prolo Scale produces results that can be compared with literature reports over many years6,7,40,47. Fusion success within the interbody space was defined by previous published parameters of radiographic fusion assessment of Brantigan and Steffee45. Two observers evaluated the fusion from Ferguson anterior-posterior (AP) (parallel to the vertebral endplates), and lateral radiographs. A level was regarded as fused if there was radiographic evidence of bone bridging the disc space without lucency. If lucency was seen to extend across the cage, the level was considered not fused. The posterior-lateral fusion was graded by the method of Lenke et al29. Each level and each side was judged individually. Continuous bridging fusion mass on at least one side was considered fusion success at that level. For patients undergoing multiple-level fusion, all surgically treated segments must be fused for the patient to be considered a fusion success.
The 44 patients in this study included 21 men (47.4%), age 39-78, mean 57, and 23 women (52.3%), age 51-83, mean 67. Of 44 patients, four were deceased and six could not be found. The remaining 34 (77%) included 15 men (71.4%) and 19 women (82.6%). The interval from previous surgery to the index ASD surgery was 79 months (7-377 months). The length of follow-up from the index ASD surgery was 77 months (36-112 months). The number of levels fused at the index ASD surgery was 1 level in 13, 2 levels in 19, and multiple levels in 12.
30 of 34 (88%) patients achieved clinical success. Excellent scores (17-20) were achieved in 5 (15%), good scores (13-16) in 6 (18%), fair scores in 19 (56%) and poor in 4 (12%). The four patients with poor scores did not achieve clinical success. One required two augmentations of the posterior-lateral fusion.
Initially after the index surgery and before additional surgical augmentation of the PLF, fusion success graded radiographically was 91% and pseudarthrosis was seen in 3/34 9%. The fusion success rate was ultimately 100%. The three pseudarthrosis patients required PLF augmentations in order to heal.
The UCLA grades identified the worst adjacent degenerative level before the index adjacent level fusion averaged 1.3 (range 1-3) and at last follow-up was average 2.4 (range 1-4) with an average change of 1 (range 0-2). There were 3 grade 4 patients that had progressed from grade 1 or 2 to grade 4 by last follow-up. The lumbar lordosis averaged 45 degrees (18-78 degrees). There were 14 (32%) with final lordosis below 40 degrees (average 30 degrees (18-39 degrees). The sagittal alignment was mildly abnormal in three (1.5 cm off plumb) and greater than 20 degrees angulation from L1-S1 in 2 patients.
Of 44 patients, there were 22 further operations in 20 patients (38%). One early revision for a displaced cage was required. 11 (18%) had extension of fusion to adjacent level and 19 (43%) had removal of pedicle screws.
One device related complication required removal of a retropulsed cage. There were no pedicle screw complications. Non-device related complications included one dural tear, two seromas that were culture negative, each requiring one debridement and primary closure, and one definite infection that required multiple debridements, long term antibiotics, and was reported to have hematogenously seeded the patient’s hip, resulting in a girdlestone procedure.
Adjacent segment deterioration after lumbar fusion is of great concern and it accounts for a substantial amount of revision spine surgery. Although some of the progression of degeneration may be attributed to natural processes of aging, this process is influenced by the abnormal stresses that occur following lumbar fusion3,11,13,14,48,50. Natural deterioration risk factors for segment degeneration have been well described, including laminar inclination, facet sagittalization and tropism, increased age, osteoporosis, female gender and post-menopausal state5,13,19,33,51. Several studies have found no statistical relationship to the length of fusion or the presence of degeneration of the adjacent segment prior to the surgery14,35,36. Park concluded radiographic asymptomatic adjacent segment disease is common but does not predict poor outcomes36. Others have related increased incidence of ASD with complete laminectomy, multiple fusion levels, severe degenerative spondylosis, facet trophism or violation by pedicle screw, failure of restoration of sagittal balance and lordosis, and extension of fusion to the sacrum15,17,24,25,43. At 5-10 years following lumbar fusion radiographic adjacent segment deterioration is commonly 30-60% and clinically symptomatic adjacent segment disease is 10-20%6,14,30,32.
Biomechanical studies have shown that lumbar fusion produces increased motion and increased intradiscal pressure at the adjacent levels9-11,18,27,31,41,44,49,52Two clinical radiographic analyses of posterior fusion patients also has shown an increase in mobility of the free segments above a fusion 13,46, while Pellise et al found no change at two adjacent motion segments37. The importance of sagittal realignment and maintenance of lordosis during fixation have been documented in clinical studies, and clinical experience suggests that lumbar fusion in a nonanatomic sagittal alignment can cause a deleterious effect at the adjacent level1,34.
Results of Surgical Treatment
Whitecloud et al. reported 58% clinical success and 83% fusion success with noninstrumented posterior-lateral fusion53. Most had only modest improvement in pain with continued need for narcotic medication53. Schlegel et al. reported 70.3% clinical success, 89.2% fusion success with noninstrumented posterior-lateral fusion, and 19% required further surgery 42. Phillips et al. reported 80.8% clinical success, 86.4%, fusion success with non-instrumented posterior-lateral fusion, and 27% re-operation rate 39. Chen et al. reported successful clinical success in 76.9% and fusion success in 94.9% of instrumented posterior-lateral fusion 8. Bertagnoli et al. report clinical success at 86% for treatment of ASD with artificial disc4
A comparison of clinical success in this ASD group (88.2%) was not statistically different to the Brantigan et al IDE study (85.9%)7 and 10 year follow-up (87.9%)6 as well as Chen et al (76.9%)8, Schlegel et al(70.3%)42, and Phillips et al (80.8%)39, although this ASD group had significantly more poor and fair Prolo scores compared to those of both of the Brantigan IDE (chi square= 20.49, pvalue=0.0216) and 10 year follow-up (chi square= 4.276,p-value =0.0387).
The fusion success was statistically significantly less in the ASD group (91.2%) than in the Brantigan IDE study (98.9%) (chi square= 7.3497, p-value=0.0067) but was not significantly different compared to the 10 year follow-up (96.7%) (chi square= 0.8199, p-value=0.3652) or to series from the literature that used non-instrumented fusion and Chen et al (94.9%) who used instrumented fusion.
Need for further surgery
Further surgery (38%) was statistically less compared to the Brantigan 10 year follow-up (69%) (chi square= 6.6656, p-value=0.0098) and significantly greater than Chen et al (5%) (chi square= 12.1952, p-value=0.0005), but not different than the Brantigan IDE (46%), and Phillips et al. (27.3%).
Adjacent segment deterioration should be considered a long term complication of lumbar fusion36. Clinical management of symptomatic ASD is difficult. When surgical treatment is elected for ASD, PLIF will result in clinical and fusion success similar to the original lumbar fusion.
Table 1: University of California at Los Angeles Disc degeneration grading scale15
*The assigned grade was based on the most severe radiographic finding evident on plain radiographs. These categories are mutually exclusive when used for grading. Patients were rated on the basis of the worst category satisfied. + = present, - = absent, and ± = either present or absent.
1. Akamaru T, Kawahara N, Tim Yoon S, et al. Adjacent segment motion after a simulated lumbar fusion in different sagittal alignments: a biomechanical analysis. Spine 2003;28:1560-6.
2. Aota Y, Kumano K, Hirabayashi S. Postfusion instability at the adjacent segments after rigid pedicle screw fixation for degenerative lumbar spinal disorders. J Spinal Disord 1995;8:464-73.
3. Axelsson P, Johnsson R, Stromqvist B. The spondylolytic vertebra and its adjacent segment. Mobility measured before and after posterior-lateral fusion. Spine 1997;22:414-7.
4. Bertagnoli R, Yue JJ, Fenk-Mayer A, et al. Treatment of symptomatic adjacent-segment degeneration after lumbar fusion with total disc arthroplasty by using the prodisc prosthesis: a prospective study with 2-year minimum follow up. J Neurosurg Spine 2006;4:91-7.
5. Boden SD, Riew KD, Yamaguchi K, et al. Orientation of the lumbar facet joints: association with degenerative disc disease. J Bone Joint Surg Am 1996;78:403-11.
6. Brantigan JW, Neidre A, Toohey JS. The Lumbar I/F Cage for posterior lumbar interbody fusion with the variable screw placement system: 10-year results of a Food and Drug Administration clinical trial. Spine J 2004;4:681-8.
7. Brantigan JW, Steffee AD, Lewis ML, et al. Lumbar interbody fusion using the Brantigan I/F cage for posterior lumbar interbody fusion and the variable pedicle screw placement system: two-year results from a Food and Drug Administration investigational device exemption clinical trial. Spine 2000;25:1437-46.
8. Chen WJ, Lai PL, Niu CC, et al. Surgical treatment of adjacent instability after lumbar spine fusion. Spine 2001;26:E519-24.
9. Chow DH, Luk KD, Evans JH, et al. Effects of short anterior lumbar interbody fusion on biomechanics of neighboring unfused segments. Spine 1996;21:549-55.
10. Cunningham BW, Kotani Y, McNulty PS, et al. The effect of spinal destabilization and instrumentation on lumbar intradiscal pressure: an in vitro biomechanical analysis. Spine 1997;22:2655-63.
11. Dekutoski MB, Schendel MJ, Ogilvie JW, et al. Comparison of in vivo and in vitro adjacent segment motion after lumbar fusion. Spine 1994;19:1745-51.
12. Fritsch EW, Heisel J, Rupp S. The failed back surgery syndrome: reasons, intraoperative findings, and long-term results: a report of 182 operative treatments. Spine 1996;21:626-33.
13. Frymoyer JW, Hanley EN, Jr., Howe J, et al. A comparison of radiographic findings in fusion and nonfusion patients ten or more years following lumbar disc surgery. Spine 1979;4:435-40.
14. Ghiselli G, Wang JC, Bhatia NN, et al. Adjacent segment degeneration in the lumbar spine. J Bone Joint Surg Am 2004;86-A:1497-503.
15. Ghiselli G, Wang JC, Hsu WK, et al. L5-S1 segment survivorship and clinical outcome analysis after L4-L5 isolated fusion. Spine 2003;28:1275-80; discussion 80.
16. Gillet P. The fate of the adjacent motion segments after lumbar fusion. J Spinal Disord Tech 2003;16:338-45.
17. Greiner-Perth R, Boehm H, Allam Y, et al. Reoperation rate after instrumented posterior lumbar interbody fusion: a report on 1680 cases. Spine 2004;29:2516-20.
18. Ha KY, Schendel MJ, Lewis JL, et al. Effect of immobilization and configuration on lumbar adjacent-segment biomechanics. J Spinal Disord 1993;6:99-105.
19. Hambly MF, Wiltse LL, Raghavan N, et al. The transition zone above a lumbosacral fusion. Spine 1998;23:1785-92.
20. Hilibrand AS, Robbins M. Adjacent segment degeneration and adjacent segment disease: the consequences of spinal fusion? Spine J 2004;4:190S-4S.
21. Ishihara H, Osada R, Kanamori M, et al. Minimum 10-year follow-up study of anterior lumbar interbody fusion for isthmic spondylolisthesis. J Spinal Disord 2001;14:91-9.
22. Kim SS, Michelsen CB. Revision surgery for failed back surgery syndrome. Spine 1992;17:957-60.
23. Kumar MN, Jacquot F, Hall H. Long-term follow-up of functional outcomes and radiographic changes at adjacent levels following lumbar spine fusion for degenerative disc disease. Eur Spine J 2001;10:309-13.
24. Lai PL, Chen LH, Niu CC, et al. Effect of postoperative lumbar sagittal alignment on the development of adjacent instability. J Spinal Disord Tech 2004;17:353-7.
25. Lai PL, Chen LH, Niu CC, et al. Relation between laminectomy and development of adjacent segment instability after lumbar fusion with pedicle fixation. Spine 2004;29:2527-32; discussion 32.
26. Lee CK. Accelerated degeneration of the segment adjacent to a lumbar fusion. Spine 1988;13:375-7.
27. Lee CK, Langrana NA. Lumbosacral spinal fusion. A biomechanical study. Spine 1984;9:574
1. Lehmann TR, Spratt KF, Tozzi JE, et al. Long-term follow-up of lower lumbar fusion patients. Spine 1987;12:97-104.
2. Lenke LG, Bridwell KH, Bullis D, et al. Results of in situ fusion for isthmic spondylolisthesis. J Spinal Disord 1992;5:433-42.
3. Miyakoshi N, Abe E, Shimada Y, et al. Outcome of one-level posterior lumbar interbody fusion for spondylolisthesis and postoperative intervertebral disc degeneration adjacent to the fusion. Spine 2000;25:1837-42.
4. Nagata H, Schendel MJ, Transfeldt EE, et al. The effects of immobilization of long segments of the spine on the adjacent and distal facet force and lumbosacral motion. Spine 1993;18:2471-9.
5. Nakai S, Yoshizawa H, Kobayashi S. Long-term follow-up study of posterior lumbar interbody fusion. J Spinal Disord 1999;12:293-9.
6. Noren R, Trafimow J, Andersson GB, et al. The role of facet joint tropism and facet angle in disc degeneration. Spine 1991;16:530-2.
7. Oda I, Cunningham BW, Buckley RA, et al. Does spinal kyphotic deformity influence the biomechanical characteristics of the adjacent motion segments? An in vivo animal model. Spine 1999;24:2139-46.
8. Okuda S, Iwasaki M, Miyauchi A, et al. Risk factors for adjacent segment degeneration after PLIF. Spine 2004;29:1535-40.
9. Park P, Garton HJ, Gala VC, et al. Adjacent segment disease after lumbar or lumbosacral fusion: review of the literature. Spine 2004;29:1938-44.
10. Pellise F, Hernandez A, Vidal X, et al. Radiologic assessment of all unfused lumbar segments
7.5 years after instrumented posterior spinal fusion. Spine 2007;32:574-9.
1. Penta M, Sandhu A, Fraser RD. Magnetic resonance imaging assessment of disc degeneration 10 years after anterior lumbar interbody fusion. Spine 1995;20:743-7.
2. Phillips FM, Carlson GD, Bohlman HH, et al. Results of surgery for spinal stenosis adjacent to previous lumbar fusion. J Spinal Disord 2000;13:432-7.
3. Prolo DJ, Oklund SA, Butcher M. Toward uniformity in evaluating results of lumbar spine operations. A paradigm applied to posterior lumbar interbody fusions. Spine 1986;11:601-6.
4. Quinnell RC, Stockdale HR. Some experimental observations of the influence of a single lumbar floating fusion on the remaining lumbar spine. Spine 1981;6:263-7.
5. Schlegel JD, Smith JA, Schleusener RL. Lumbar motion segment pathology adjacent to thoracolumbar, lumbar, and lumbosacral fusions. Spine 1996;21:970-81.
6. Shah RR, Mohammed S, Saifuddin A, et al. Radiologic evaluation of adjacent superior segment facet joint violation following transpedicular instrumentation of the lumbar spine. Spine 2003;28:272-5.
7. Shono Y, Kaneda K, Abumi K, et al. Stability of posterior spinal instrumentation and its effects on adjacent motion segments in the lumbosacral spine. Spine 1998;23:1550-8.
8. Steffee AD, Brantigan JW. The variable screw placement spinal fixation system. Report of a prospective study of 250 patients enrolled in Food and Drug Administration clinical trials. Spine 1993;18:1160-72.
9. Stokes IA, Wilder DG, Frymoyer JW, et al. 1980 Volvo award in clinical sciences. Assessment of patients with low-back pain by biplanar radiographic measurement of intervertebral motion. Spine 1981;6:233-40.
10. Stromberg L, Toohey JS, Neidre A, et al. Complications and surgical considerations in posterior lumbar interbody fusion with carbon fiber interbody cages and Steffee pedicle screws and plates. Orthopedics 2003;26:1039-43.
11. Torgerson WR, Dotter WE. Comparative roentgenographic study of the asymptomatic and symptomatic lumbar spine. J Bone Joint Surg Am 1976;58:850-3.
12. Umehara S, Zindrick MR, Patwardhan AG, et al. The biomechanical effect of postoperative hypolordosis in instrumented lumbar fusion on instrumented and adjacent spinal segments. Spine 2000;25:1617-24.
13. Untch C, Liu Q, Hart R. Segmental motion adjacent to an instrumented lumbar fusion: the effect of extension of fusion to the sacrum. Spine 2004;29:2376-81.
14. Vanharanta H, Floyd T, Ohnmeiss DD, et al. The relationship of facet tropism to degenerative disc disease. Spine 1993;18:1000-5.
15. Weinhoffer SL, Guyer RD, Herbert M, et al. Intradiscal pressure measurements above an instrumented fusion. A cadaveric study. Spine 1995;20:526-31.
16. Whitecloud TS, 3rd, Davis JM, Olive PM. Operative treatment of the degenerated segment adjacent to a lumbar fusion. Spine 1994;19:531-6.
17. Wiltse LL, Winter RB. Terminology and measurement of spondylolisthesis. J Bone Joint Surg Am 1983;65:768-72.